Views: 7 Author: Site Editor Publish Time: 2020-03-13 Origin: Site
The packed ultrasonic heat meter should not be directly affected by rain, frost and fog during transportation. It should be placed upwards according to the signs and should not be damaged by crushing or impact. The environmental conditions of transportation shall be implemented in accordance with the provisions of the JB / T9329 standard.
Storage conditions: The product pad should be at least 0.3m from the ground, at least 1m from the four walls, and not less than 2m from the heating equipment. The ambient temperature is 5 ℃ ~ 55 ℃, and the relative humidity is <80%. The warehouse should be free of acids, alkalis, flammable, explosive, toxic and other chemicals. Also avoid other corrosive gases and objects, and avoid strong electromagnetic fields and direct sunlight.
After the ultrasonic thermometer is installed and used, it should be checked regularly. It is used to prevent the working state of the thermometer and the battery voltage from working normally.
When the flow is significantly reduced and the heating is affected, the filter should be drained and cleaned.
When the ultrasonic calorimeter works for 6 years. After more than 6 years or the display data is not clear or the voltage shows "V", the battery should be replaced by a professional technician. This can avoid affecting the normal operation of the calorimeter.